Hydraulic Relief Valve,Cartridge Relief Valve,Crossover Relief Valve,Hydraulic Pressure Relief Valve Dongguan Jinzhuang Hydraulic Technology Co., Ltd. , https://www.jinzhuanghydraulic.com
The tester bench is made of stainless steel materials; The surface is treated with polishing.
Clamp equipment and drain machine is controlled by four-way reversing valve; The water flow rate is adjusted by copper case stainless steel core ball valve, it has high accuracy, the structure is compaction, long life, easy operate and maintain.
2. Main technical parameter
2.1 The tester bench accuracy: 0.2
2.2 Tested water meter size: DIN15, DIN20, DIN25, DIN40, DIN50
2.3 Technical criteria for water meter testing: Used for full capability test or produce andmaintenance
2.4 Instantaneous flow index gauge is adopted glass rotor flowmeter group, the actual
Measurement accuracy not less than 2.5%
The detail type and specification as following:
LZB-80 7-31.5m3/h (Additional for 15-50mm tester)
2.5 Working volume container cubage:
15-50mm, (10L+10L)20L, (100L+200L)300L.
2.6 Using water source working pressure: 0.25-0.4MPa
2.7 Clamp equipment flex distance is:
2.8 Appearance size
2750Ã— 1200Ã— 2100 (in series meter tester)
3. Structure and working principle
The water meter tester bench is composed by work desk, clamp equipment, front and back straight pipe, flowmeter, standard volume container, drain water valve.
After the water from the steady sources come into the enter place of tester bench, the water will come into the tested meter in straight pipe, and then come into the choosed standard volume container, compare the reading number between tested meter and standard volume container, and get the error of testing water meter.
Hydraulic Relief Valve,Cartridge Relief Valve,Crossover Relief Valve,Hydraulic Pressure Relief Valve
Dongguan Jinzhuang Hydraulic Technology Co., Ltd. , https://www.jinzhuanghydraulic.com